RADAR TEST SET
OPERATION UNDER USUAL CONDITIONS - Continued
Table 6-7. Interrogator/TCAS Manual Tests - Continued
From "Inter Test Manual?"
In Direct Connect mode, "Mode 1
screen, go to Mode 1.
...Direct...." screen is shown for the Mode
In Radiated mode, "Mode 1 .........XX ft."
screen is shown for the Mode 1 testing.
Tests can be repeated by pressing S2.
The operator can go to the Interrogator
Temp Setup by pressing S2 for at least 3
seconds and releasing. Refer to
The operator can view parametric data by
pressing S3. Subsequent pushes of S3
displays additional parametric screens,
ultimately cycling back to the first page.
Continue to sequence through
Press S1 to go through each
The operator can back up to a previous
the individual test modes (M2,
test in the sequence by holding down the
M3, MC, Mode 4A(1), Mode
S1 button for at least 3 seconds and
4B(1), and Mode S).
Store results (if desired).
Pressing the S2 button when
Once the test sequence has been
in the Save Test Data screen
completed, the operator is given the
provides YES and No options.
option to save the test data to memory.
If saved, the test number is displayed. In
Press S1 to execute.
order to correlate the test results with the
test aircraft the operator must record the
test number and aircraft tail number. The
data can be downloaded to a PC when
testing has been completed.
If desired, download and
view/print stored results using