RADAR TEST SET
ASSEMBLY AND PREPARATION FOR USE - Continued
Verify the Test Set display "BIT Test: PASS" at the end of the BIT sequence. If a problem with the Test
Set operation is indicated by "BIT Test: FAIL", refer to Troubleshooting Section TS007 (WP0017 00). If
the "BIT Test: PASS" screen is displayed, the Test Set is ready for use.
BIT Test: PASS
SAVE DIRECT CONNECT CABLE CAL FACTOR.
To improve measurement accuracy during direct connect testing, the Test
Set can compensate for the loss of the direct connect cable. The direct
connect cable loss is marked on a tag attached to the cable. The cable cal
factor is stored in the Test Set even when the battery is removed.
To initially store the cable cal factor in a new Test Set or update the cal factor if the direct cable is replaced,
utilize the bench utility (PN: 60-0694-B2). The cable cal factor is marked on the tag attached to the direct
connect cable. Refer to WP0008 00 utilizing the bench utility for downloading and viewing the direct connect
cable cal factor in the Test Set.